The setup in Figure 2 illustrates the impact of poor quality connectors. In this image, the TDR is deskewed with no cables attached to set the reference plane at the TDR instrument front panel. A high-quality semi-rigid coax is connected to the front panel while a thru connector is used to attach a second high-quality cable. Finally, the second cable is attached to a PCB 50Ohm test trace with a low-quality PCB edge connector.
top of page
bottom of page