We bought a couple of those B-Amp12 to boost the signal level for doing PDN measurements and I had a question about connections for the amp.
The amp has connections for Ch1 and Ch2, If I connect those to the Omicron, how/where do I connect the other end of the P2102A probe?
Does the Bode 100 SW recognize its there and let me turn up the signal to 25dbm?
Do I need to Ch1 on the Omicron?
I checked your website and was looking for a doc using the source power amp, but couldn’t find one.
For doing cal during open a load I am guessing this really wouldn’t change the level since I have found that the max level the Omicron can handle when doing probe cal is around -7dbm, but this would give us considerably more dynamic range when it comes to a SHORT cal correct (which is the calibration that really matters measuring low impedance PDNs anyways?)